Realtime datalog test from Raspberry Pi. AD584KH 5V measured with K2002-u1, K2002-u2, 3458A
Please wait for all data to get loaded. It may take 10-30 seconds in worst case.
If some graphs (should be 4) are missing, try to reload page.
Warning, CAVEAT! Some data points in CSV's are missing, but are interpolated on graphs. These interpolated segments do not have REAL DATA! You can see them as very smooth sections on graph. Beware.
Download CSV files for full datalog details
CSV-data from Keithley 2001 New (NPLC50)
CSV-data from Keithley 2002 Old (NPLC50)
CSV-data from HP 3458A (NPLC100)
Workflow:* RPI accesses DMMs via GPIB and reads data.
* Data is formatted by python script into DSV string and written/append to file on FTP
* DSV-file is visible publicly on https://xdevs.com/datashort/
* D3.js plots SVG graph online :)