CalMachine KX2 Temperature box test, GPIB code running on Raspberry Pi 3 + NI GPIB-USB-HS FC
TC Result = +0.06 ppm/K peak, TC0.05 window point ends at +54 °CTrim resistor = 305K
PID settings: KP=140, KI=0.2, KD=1.2, Max TEC voltage = 10.5V, Max TEC Current = 1A, YSI 44006 current = 100E-6 A
LTZ CM1 KX2 REF = 7.157404237 VDC;
Workflow:* RPI accesses DMMs via GPIB and reads data.
* Data is formatted by python script into DSV string and written/append to file on FTP
* DSV-file is visible publicly on https://xdevs.com/datashort/
* D3.js plots SVG graph online :)