SR104 arrival verify over 2002-4
Reference levels: 10000.0011 Ω at +23.34 °C, well bottom temp with 100 µA F8508A TrueOhm bridge to Rear STD SR104
Calibration data from year 2016.
Workflow:* RPI accesses DMMs via GPIB and reads data.
* Data is formatted by python script into DSV string and written/append to file on FTP
* DSV-file is visible publicly on https://xdevs.com/datashort/
* D3.js plots SVG graph online :)