- Exterior, controls and initial condition
- Internal construction, design features
- Diagnostics and secret menu
- Performance test
- Application example, measuring resistance
This current source, build on the platform of 11W four quadrant SMU is common tool in scientific labs and often used for materials study and evaluation. Common use is delta-measurements for resistance using another Keithley instrument – Model 2182/2182A nanovoltmeter. Sensitivity and offset cancellation functions of such tandem allow easy and accurate measurements of very low resistances and shunts. 6221 can source very wide range of currents, from sub-nA (lowest range 1nA) to quite powerful 100mA, while maintaining up to ±105V of compliance voltage.
However this versatility comes at price, around 5520$ USD new for bare Keithley Model 6221. I’ve got unit for this article for bit over $1300 USD.
Today market niche of this instrument is overlapping with other precision SMU, such as own Keithley 2600 series and competitor Keysight B2900A series, however these cost bit more.
Redistribution and use of this article or any files referenced in it, in source and binary forms, with or without modification, are permitted provided that the following conditions are met:
- Redistribution of article must retain the above copyright notice, this list of conditions, link to this page (https://xdevs.com/article/kei6221/) and the following disclaimer.
- Redistribution of files in binary form must reproduce the above copyright notice, this list of conditions, link to this page (https://xdevs.com/article/kei6221/), and the following disclaimer in the documentation and/or other materials provided with the distribution, for example Readme file.
All information posted here is hosted just for education purposes and provided AS IS. In no event shall the author, xDevs.com site, or any other 3rd party, including Keithley Instruments or Tektronix be liable for any special, direct, indirect, or consequential damages or any damages whatsoever resulting from loss of use, data or profits, whether in an action of contract, negligence or other action, arising out of or in connection with the use or performance of information published here.
If you willing to contribute or add your experience regarding any test instruments or provide any extra information, such as firmware dumps, internal photographs, you can do so following these simple instructions
Exterior, controls and initial condition
Internal construction, design features
Diagnostics and secret menu
No repair was required.
Calibration procedure in manual calls for HP 3458A 8½-digit multimeter and Keithley 6487 picoammeter. These ones I have, with little deviation by using Keithley 6485 instead of 6487. DMM is used to verify current ranges from 100mA to 100nA and picoammeter take care of lowest 10 nA and 1 nA ranges.
|Test condition||3458A reading||Error from DMM||6221 Specification (1 year)||Result|
|Test DCI 100 mA||99.9816 mA Datalog||-184.0 ppm||±1500 ppm||PASS|
|Test DCI 10 mA||10.00076 mA Datalog||+76 ppm||±1500 ppm||PASS|
|Test DCI 1 mA||1.000038 mA Datalog||+38 ppm||±600 ppm||PASS|
|Test DCI 100 µA||100.0056 µA Datalog||+56 ppm||±600 ppm||PASS|
|Test DCI 10 µA||9.99984 µA Datalog||-16 ppm||±600 ppm||PASS|
|Test DCI 1 µA||0.999848 µA Datalog||-152 ppm||±1100 ppm||PASS|
|Test DCI 100 nA||99.9884 nA Datalog||-116 ppm||±3100 ppm||PASS|
Application example, measuring resistance
Modified: July 2, 2017, 7:49 p.m.