Airbath test GPIB code running on Pi3 + NI GPIB-USB-HS FCSEP19 check,
Verification of xDevs.com QVR 1 BG modules
Test against BelGIM : MAY.22.2020, batt-powered QVR1 , 9.9909193 VDC
Workflow:* RPI accesses DMMs via GPIB and reads data.
* Data is formatted by python script into DSV string and written/append to file on FTP
* DSV-file is visible publicly on https://xdevs.com/datashort/
* D3.js plots SVG graph online :)