Test of resistance box 95KOhm in ambient. 3458A@NPLC100 OCOMP ON with 5s delay
TCBOX is -1.314 ppm/K
Resistance: 94996.5057 Ohm, -36.7821 ppm from 95K
Result CSV-data from HP 3458A
Workflow:* RPI accesses DMMs via GPIB and reads data.
* Data is formatted by python script into DSV string and written/append to file on FTP
* DSV-file is visible publicly on https://xdevs.com/datashort/
* D3.js plots SVG graph online :)