Test of Time 9823 100 output in ambient. 3458A@NPLC100 OCOMP ON with 5s delay
TCBOX is +1.0 ppm/K after initial soak 1 day
Resistance: 99.96981 Ohm, -301.9 ppm from 100 Ohm
Result CSV-data from HP 3458A
Workflow:* RPI accesses DMMs via GPIB and reads data.
* Data is formatted by python script into DSV string and written/append to file on FTP
* DSV-file is visible publicly on https://xdevs.com/datashort/
* D3.js plots SVG graph online :)