Test of Time 9823 10M output in ambient. 3458A@NPLC100 OCOMP ON with 5s delay
TCBOX is +1.93 ppm/K after initial soak 1 day
Resistance: 10010405.504 Ohm, +1040.5 ppm from 10M Ohm
Result CSV-data from HP 3458A
Workflow:* RPI accesses DMMs via GPIB and reads data.
* Data is formatted by python script into DSV string and written/append to file on FTP
* DSV-file is visible publicly on https://xdevs.com/datashort/
* D3.js plots SVG graph online :)