Test of KX LTZ1000A reference, various box temperatures, battery powered. Readings 3458A@NPLC200
Please wait for all data to get loaded. It may take 10-30 seconds in worst case.
Result CSV-data from HP 3458A
Workflow:* RPI accesses DMMs via GPIB and reads data.
* Data is formatted by python script into DSV string and written/append to file on FTP
* DSV-file is visible publicly on https://xdevs.com/datashort/
* D3.js plots SVG graph online :)