TECKIT test B2, VPG inboundHPA = 4.53K VHP103ZT R4
6581T = 3K VHP103ZT R5
K2002-6 = 3K VHP102ZT R6
PID settings: KP=140, KI=0.1, KD=0.9, Max TEC voltage = 10.5V, Max TEC Current = 1.3A, YSI 44006 current = 100E-6 A
Workflow:* RPI accesses DMMs via GPIB and reads data.
* Data is formatted by python script into DSV string and written/append to file on FTP
* DSV-file is visible publicly on https://xdevs.com/datashort/
* D3.js plots SVG graph online :)