- Manual references
- Initial inspection and basic disassembly
- Update from reader on newer DMM6500
Common industry-known 6½
digit DMM from Keithley, Model 2000 just got refreshed with new -blingy fancy design, as Tektronix/Keithley DMM6500. However this is not just a simple facelift, but actually meter redesigned from scratch. Tektronix released this DMM6500 in April 2018. Let’s see what makes it tick.
Other brand competitors for this meter would be old Fluke 8846A and Keysight 3446x series.
|DC Voltage||±10mV, ±100mV, ±1V, ±10V, ±100V||Best 1 Year Specification: ±25 ppm + 2|
|AC Voltage||±100mV, ±1V, ±10V||Best 1 Year Specification: ±25 ppm + 2|
|DC Noise spec||6 nV & 25s NPLC5, FILT, 10mV, 2.5µV & 1s NPLC1, FILT, 10V|
|Resistance 4W||1 Ω – 100 MΩ|
|DC Current||10, 100 µA, 1,10,100 mA, 1,3,10 A|
|AC Current||100 µA, 1,10,100 mA, 1,3,10 A|
|Capacitance||1,10,100 nF, 1,10,100 µF|
|Digitizer||Up to 1 MSPS for V/I|
|Diode||10V clamp (10,100 µA, 1,10 mA)|
|Temperature||-200°C to +1820°C, Type J,K,N,T,E,R,S,B||Accuracy: ±0.2 °C, 0.001°C resolution|
|ADC linearity||±0.8ppm of reading + 0.5ppm of range|
|Input impedance||>10 GΩ for 10mV-10V ranges|
Table 1: Model DMM6500 measurement functions
Let’s also look on other market competitors:
|Specification||Tektronix/Keithley DMM6500||Keithley 2000||Keysight 34465A||Keithley 2001||Keithley DMM7510||Keithley 2002|
|Resolution||6½ digits||7½ digits||8½ digits|
|ACAL function||No||No||Yes||ACV only||Yes||ACV only|
|Form-factor||Half-rack 2U 19” instrument|
|MSRP||$1140 USD||$1140 USD||$1430 USD||$4600 USD||$4355 USD||$6150 USD|
Table 1: Comparison summary to other popular meters on market
Redistribution and use of this article or any images or files referenced in it, in source and binary forms, with or without modification, are permitted provided that the following conditions are met:
- Redistribution of article must retain the above copyright notice, this list of conditions, link to this page (/review/dmm6500/) and the following disclaimer.
- Redistribution of files in binary form must reproduce the above copyright notice, this list of conditions, link to this page (/review/dmm6500/), and the following disclaimer in the documentation and/or other materials provided with the distribution, for example Read-me file.
All information posted here is hosted just for education purposes and provided AS IS. In no event shall the author, xDevs.com site, or any other 3rd party, including Keithley or Tektronix be liable for any special, direct, indirect, or consequential damages or any damages whatsoever resulting from loss of use, data or profits, whether in an action of contract, negligence or other tortuous action, arising out of or in connection with the use or performance of information published here.
If you willing to contribute or add your experience regarding instruments repairs or provide extra information, you can do so following these simple instructions
As usual, all photos are clickable for high-resolution version.
Unfortunately, as with nearly all modern instruments, there is no more schematics or theory of operation information available in public manuals for Model DMM6500. Even service manual or calibration manual is not available at the time of writing this review.
Initial inspection and basic disassembly
Rear side quite busy with connector and option ports. Communication option feature GPIB card. Smaller populated card in center is scanner option. Old scan cards like 2001-TCSCAN and 2000-SCAN are fully supported.
There is no hardware power switch for mains power, so meter is always have stand-by power present.
Probes are typical for benchtop meter.
“We are in like Flynn”. Large mainboard with many slots and sections contain all the meter’s circuitry.
Transformer is rigidly fixed to the chassis frame, which on early units caused some unwanted buzzing noise. Later units, according to Keithley, already addressed this.
Not much components on this side, mostly passives with few chips.
Input DCI function is protected by large fuse in holder. Mechanical front/rear switch also visible on the corner, with rest of input protection parts.
Front panel PCBA busy with BGA chips, as big fancy LCD need lot of juice to process and refresh the images quickly. Two Freescale/NXP chips in center are MPC5125 32-bit microprocessors with e300c4 Power Architecture® core, display interface, 10/100 MAC, USB and all usual interfaces.
Not sure why need two processors. Perhaps once is handling display/front panel functionality only, while the second actually runs meter firmware/functions? Lattice LFE5U is ECP5 FPGA, featuring 24K LUT elements and about a megabit of onboard memory. It supports lot of high-speed interfaces, and may be the one responsible of driving LVDS LCD.
Mainboard PCB is designed with six layer metal interconnect. This is clear, thanks to edge PCBA layer marker.
Laser-trimmed hermetic film resistor visible on the left side. That is high-voltage divider for DCV/ACV ranges. Very common in Fluke meters and calibrators.
Si8631 and few opto’s take care of the insulation between outguard and inguard sides. Typical design for benchtop DMM, as these meters need to have front-end floating.
Small 40 × 40 mm DC fan located next to transformer. Hopefully this fan does not get too noisy after few years of use.
Transformer shares same assembly block with inductor. Inductor is used for PFC correction for the mains input.
Rear terminal jack with IDC cable? That’s not really what we expected to see in 6½-digit meter with 10 µA current range.
Update from a reader on his DMM6500
Robert S. sent us some more photos of Keithley DMM6500 inner workings. Thank you for contribution.
This DMM was purchased new in March 2021 and have few differences inside:
There is a mod wire soldered from U32.3 to K7.2-3. Also Keithley put a silkscreen labels on board to specify when K5 will be populated or not, depends on option model.
This DMM was assembled around 46th week of 2020. LM399A reference with special partnumber for Keithley was cooked around 10th week of 2020.
Sadly we can learn that transformer still makes the same horrible hum, no enhancements only the label has been styled. Transformer hum issues were supposed to be fixed since original DMM6500 release to market.
To be updated.
Discussion about this article and related stuff is welcome in comment section or at our own IRC chat server: irc.xdevs.com (standard port 6667, channel: #xDevs.com). Web-interface for access mirrored on this page.
Modified: March 20, 2021, 7:02 p.m.